I. Product introduction:
The electric comprehensive surgical table is for use in comprehensive surgeries in thoracic-abdominal surgery, surgery, cerebral surgery, ophthalmology, E.N.T., obstetrics and gynecology, urology and other departments.
II. Unique product advantages:
1. The table top is made of high-strength and X-ray radio-transparent composite materials to meet the need of clinical X-ray imaging.
2. The mattress is made of slow rebound memory foam, which is comfortable and beautiful and can evenly bear the pressure of the load. Its advantages include corrosion resistance, acid & base resistance and breathability, and is easy to clean. The mattress can naturally shape according to the patient’s body temperature to effectively prevent bedsore.
3. The head and leg boards are detachable and the leg board can be rotated outward and folded downward manually to facilitate operations and better meet the need of various surgical positions as well as provide more flexible operating space for medical staff
4. The handhold controller uses a 24V DC power supply, and its appearance design is smooth, ergonomic, safe and reliable and comfortable to hold, providing stable and accurate operations;
5. The surgical table is made of medical-grade 304 stainless steel and its surface is specially treated with matte finishing, which looks more modern and is easier to clean;
6. The high-strength stainless-steel bottom cover has strong pressure resistance, corrosion resistance and is easy to clean;
7. The base of the surgical table can move or keep still. The bottom of the base is equipped with wheels for flexible movement and reliable immobilization;
8. The table top has an optional built-in waist bridge to facilitate biliary and kidney surgeries;
9. Main actions of the surgical table: lifting and lowering, forward & backward tilting, leftward and rightward tilting of table top and lifting of back board are controlled by an electric device composed of a controller and an electric push rod; turning of head board and leg board is controlled by a gas spring; lifting of waist board is mechanically controlled.
10. The electrically controlled part is mainly composed of a power adapter, control button box, etc. Power supply used by the surgical table: the external 220V AC is transformed into 24V DC by a power adapter to drive the motor.
III. Technical parameters:
|Length of table top||2020(± 20) mm|
|Width of table top||500(±20) mm (without side guide)|
|Height of table top (electric)||Minimum 720(± 20) - maximum 1100(± 20)mm (electric)|
|Forward and backward tilting angles of table top (electric)||Forward tilting ≥19°, backward tilting ≥15°|
|Leftward and rightward titling angles of table top (electric)||Leftward tilting ≥18°, rightward tilting ≥18°|
|Head board turning angle (manual)||Upward turning ≥46°, downward turning ≥90°|
|Back board turning angle (electric)||Upward turning ≥68°, downward turning ≥15°, detachable|
|leg board turning angle (manual)||Upward turning ≥15°, downward turning ≥90°, axle type, expandable angle 180°, detachable|
|Waist bridge lifting distance||120mm(±10)|
IV. Basic Configuration List (Single Device)
|1||Body of surgical table||Chenhong||1||Set|
|3||Motor (optional imported ones)||JIECHANG||4||sets|
|4||Anesthetic screen rack||Chenhong||1|
|9||Built-in waist bridge||Chenhong||1||Set|
|12||Product Certificate/Warranty Card||Chenhong||1||Set|
|13||Instructions for Use||Chenhong||1||Set|
Optional Configuration List (Single Device)